Step 5 :
Starting an electrostimulation session
1. Press the green start button on top of the device
2. Press the up arrow to start a new session, and confirm using the green button
3. Increase amperage until the patient begins to feel tingling, then decrease by one (1) mA using "-" button. Amperage can be set between 1 and 42mA. An alarm signal is activated at less than 5mA.
The average is generally 20 to 25mA. This sensitivity may vary between sessions.
Make sure the yellow LED is on (current indicator)